Visit us on the web at for more information: • Price Quotations • Drivers· Technical Specifications. Manuals and Documentation . Model System DMM Instruction Manual Rev. D. Manual Type: Primary User; Part Number: D. Release Date: Sep 17 View and Download Keithley DMM instruction manual online. 5 function autoranging digital multimeter. DMM Measuring Instruments pdf manual.
|Published (Last):||6 October 2011|
|PDF File Size:||20.41 Mb|
|ePub File Size:||9.14 Mb|
|Price:||Free* [*Free Regsitration Required]|
If the factory defualt value is the current B value, the following message will be displayed: When the Model completes the reading, it will tng- ger the Model to go to the next channel. Determine the center frequency and bandwidth as ex- plained in the previous application Measuring Bandwidth.
Address the Keithly to listen. However, the instrument must be allowed to warm up for at least two hours to achieve rated accuracy. Thus the dynamic measurement range of the instrument is Triggering may be done in two basic ways: Sending ID over the bus will not change the current line frequency setting, and will save that setting as the default setting.
The calibration signals can be either prompted default values or numbers entered from the front panel. Note that the instrument remains in the listener active state after the command is transmitted.
User selected values of M and B will be stored within the Model until the power is turned off unless saved by Program When any of these commands are transmitted, the instrument will send the appropriate reading the next time the instrument is addressed to talk. At 5 V 2 d resolution, the number of counts would be reduced by a factor of Powering Results with Precision and Simplicity Guide: Three time constants are used to allow the filters to settle.
For keithlet ample, if the software level is Bl, the following message will be displayed: LOCAL Note that the instrument goes into the local mode, and that operation of the front panel keys has now been restored. Through the use of the H command, the front panel programs may be entered through commands given over the bus.
The first digit is the interface select code, which is set to 7 at the factory. Note that commands are not necessarily executed in the order received; instead, they will be executed in alphabetical order.
Refer to paragraphs 2. Source Measure Unit Instruments Tech: To make sure that correct status is transmitted, the status word should be requested as soon as possible after the command is transmitted. Instrument Division Cleveland, Ohio, U. The computer first places the instru- ment in remote line 10 and then programs the SRQ mode of the instrument line Ticket page with photos Photos.
Enter the number 4 by pressing the “4” button. Increase the frequency input until the Model reads -S. Cal Locked Error A cal locked error occurs when trying to calibrate the in- strument over the bus with the front panel calibration switch in the disable position.
Keithley 196 DMM Manuals
Offset-Compensated Ohms-Offset-compensated ohms is used to compensate for voltage potentials such as thermal EMFs across the device under test.
Values sent with a wild card Translator word select options of the equivalent DDC. In the continuous trigger mode, the instrument takes a continuous series of readings. Bit 1 Buffer Full — Set when the defined buffer size is full. FOX— Single command string. Leave a Reply Click here to cancel reply.
At SVzd resolution, the number of counts would be reduced by a factor of You can download the firmware along with the calibration data using the links at the end. Factory Default Conditions At the factory, the Model is set up so that the instru- ment is configured to certain setup conditions on the initial power up.
The B command affects the source of the data. Figure shows the general format of this mask.